One of the free services we offer to customers here at EXAIR is our Efficiency Lab. In case you are not familiar here is a brief synopsis. Speak with an Application Engineer about your existing compressed air blowoff/point of use product and that you would like to know how much air it consumes. Fill out the brief survey and send the product you use in to our facility. Let us perform tests on calibrated test equipment to determine the force, flow, and noise level. We will then issue you a report that states what the EXAIR model would best be suited (if applicable) as well as how much compressed air you will be able to save. Order the recommendation and start saving money.
To do these evaluations, we have to have calibrated equipment that is reliable and capable of handling vast range of products we may receive in. For this, we could use a Digital Flowmeter, in some cases that is what has to be done due to large flow rates. For the majority of these though we go old school. We utilize a piece of equipment called a rotameter.
This is a device that is designed to measure the flow rate of a fluid within a closed tube. The inside diameter of the tube is varied which causes the float within the meter to raise or lower. They are calibrated for a specific gas at a given pressure and temperature, most are calibrated for atmospheric conditions, 14.7 psi (1.014 Bar). The meter must be mounted vertically and this is not always best suited for industrial environments.
When testing products the compressed air within the meter is pressurized which means we have to correct the reading for the given pressure, if the temperature is outside of the calibration temp then we must also perform that correction. We do this using a table provided by the manufacturer of the meter or by using the calculations shown to get exact values that may be in between the pressures in the table.
This will allow us to then multiply the Correction Factor by the meter reading and calculate our corrected flow for the point of use device at a given operating pressure and temperature.
Knowing where the values that are measured and calculated come from add validity to the reports and understanding all of the variables that go into reading like this helps to better validate the cost savings that can be seen.
In a pinch, for a field estimation, we can also use these Correction Factors and determine an approximate consumption rate of a device that has been measured at a pressure such as our cataloged 80 psig (5.5 Bar). This can often be done on the fly to help determine the flowrates currently on a system. This can be helpful when troubleshooting, giving estimated simple ROIs, and help justify results and reasons for future purchases of engineered solutions.
If you want to discuss the Efficiency Lab or any of the math behind our calculations, contact any Application Engineer, we can all help out.